Seat Belt Tension Transducer
Measuring Impact-Initiated Tesion of Seat Belt in Crash Tests
Using a strain gage, the LBT-A-20KNSA1-IDT detects the impact-initiated tension applied through the dummy to the seat belt.
|Rated Capacity||20 kN|
|Nonlinearity||Within ±2% RO|
|Rated Output||Approx. 2.5 mV/V|
|Safe Temperature||0 to 60ºC|
|Compensated Temperature||10 to 50ºC|
|Temperature Effect on Zero||Within ±0.1% RO/ºC|
|Safe Excitation||10 V AC or DC|
|Recommended Excitation||1 to 5 V AC or DC|
|Input Resistance||350 Ω ±2%|
|Output Resistance||350 Ω ±2%|
|Insulation Resistance||500 MΩ or more (With 25 VDC applied)|
|Cable||4-conductor (0.08 mm2) vinyl shielded cable, 3.2 mm diameter by 7 m long, terminated with a connector plug 3RT01-PE7M (TEDS installed)
(Shield wire is not connected to the case.)
|Weight||Approx. 130 g|
Single belt calibration (Our standand seat belt)
Note) Characteristic values are based on our calibration methods and conditions.
*We accept calibration for the supplied seat belt.