Search transducers
Select the kind to narrow the selection range
00items were found.
00items were found.
Further narrow the selection range
Test targets
 Automotive performance test
 Component test
 Crash safety test
Product image  Type  Description 

LPRC Thin Pedal Force Transducer 
Easy mounting applicable different pedal configurations and offset load provide low output errors.  
LSAAS1 Gearchange Lever Operating Force Transducer 
Minimal Error due to Different Gripping Applied to a gearchange lever for floor shift, the LSAAS1 measures 2component force generated by operating the gearchange lever. 

LBB Hand Brake Operating Force Transducer 
The LBB is a load cell to measure tension of hand brakes. It can be install to a hand brakes easily. Adopting universal joint, the transducer can measure accurately. 

Slipring Type Wheel Torque Transducer 
Measures wheel torque during driving or breaking. ●Measures the wheel torque generated by driving or braking an automobile. ●Applicable for braking and traction control measurements ●Compact & lightweight ●Optional angle detector outputs signals of rotary angle of wheel simultaneously. 

Inwheel Torque Sensor 
Applicable to braking and driving measurements  
PE30KF Engine Pressure Transducer 
The PEF is a pressure transducer for measurement of burning pressure inside engines. Rated Capacity: 3 MPa Safe Temperature: 300ºC or less for attaching screw section, and 80ºC or less for waste water 

SFAESA20 Steering Force/Angle Transducer 
This unit is thin and lightweight design, easy to mount to and remove from the steering shaft. Strain gages are used to detect steering force (torque) and equipped with strain amplifier. Rotary encoder for steering angle detection is used and enable high resolution. Dedicated indicator is also available for indication of steering force and steering angle. 

SFAFSA Steering Force/Angle Transducer 
Easy disconnection from steering shaft This unit is easy to mount to and remove from the steering shaft, and both steering force (torque) and steering angle can be output as strain. *Type that adds output for small angles (rated ±160°) output can also be manufactured. 

SFAHSA Steering Force/Angle Transducer 
Lightweight, for easy disconnection General from steering shaft Unit is lightweight, for easy mounting to and removal from the steering shaft 

LSGA Gripdesign Steering Operation Force Transducer 
To evaluate the steering operational force from hand to steering wheel 

ASDB Acceleration Transducer for Crash Test 
The ASDB is a undamped type compact & lighweight acceleration transducer for crash tests. There are two types vertical models ASDB1KV/2KV and horizontal models ASDB1KH/2KH. Rated Capacity: ±9807 to ±19613 m/s^{2} Dimensions: 10 x 4.7 x 12.5 mm Rated Output: 1.2 mV/V or more Frequency Response: DC to 4 kHz, sensitivity deviation: ±5% at 23ºC 

ASDHA Acceleration Transducer for Crash Test 
The ASDHA is a undamped type compact & lighweight acceleration transducer with a sputtertype strain gage as the sensing element. There are two types vertical (ASDHA1KV/2KV) and horizontal (ASDHA1KH/2KH). Rated Capacity: ±9807 to ±19613 m/s^{2} Dimensions: 10 x 4.7 x 12.5 mm Rated Output: 1.2mV/V or more Frequency Response: DC to 4 kHz, sensitivity deviation ±5% at 23ºC 

ASEA Damped Acceleration Transducer for Crash Test 
The ASEA is an oil damped acceleration transducer for crash tests. It is suitable for measuring vibration of car bodies, components, and dummies. Rated Capacity: ±980.7 to ±9807 m/s^{2} Dimensions: 10 x 5 x 10 mm Rated Output: 0.5 mV/V or more (±980.7 m/s^{2}), 1 mV/V or more (±4903 m/s^{2}), 2 mV/V or more (±9807 m/s^{2}) Frequency Response (23ºC): DC to 1 kHz for all models (Sensitivity deviation ±3%), 1 kHz to 2 kHz for ASEA100 (Sensitivity deviation ±10%) (±980.7 m/s^{2}), 1 kHz to 4 kHz for ASEA500/1K (Sensitivity deviation ±10%) (±4903 m/s^{2}) 

ASERA Damped Acceleration Transducer for Crash Test 
The ASERA is an oildamped type, suitable for measuring car bodies and dummies. Its cable is exchangeable by repair. Rated Capacity: ±980.7 to ±9807 m/s^{2} Dimensions: 10 x 5 x 10 mm Rated Output: 0.5 mV/V or more (±980.7 m/s^{2}), 1 mV/V or more (±4903 m/s^{2}), 2 mV/V or more (±9807 m/s^{2}) Frequency Response: (23ºC): DC to 1 kHz for all models (Sensitivity deviation ±3%), 1 kHz to 2 kHz for ASERA (Sensitivity deviation ±10%) (±980.7 m/s^{2}), DC to 1 kHz for ASERA500/1K (Sensitivity deviation ±3%) (±9807 m/s^{2}) 

ASDEA Damped Acceleration Transducer for Crash Test 
The ASDEA is an oildamped type acceleration transducer for crash tests. It is least affected from resonance. Rated Capacity: ±980.7 to ±9807 m/s^{2} Dimensions: 10 x 4.9 x 13.5 mm Rated Output: 1 mV/V or more Frequency Response: DC to 1 kHz, sensitivity deviation ±3%, 1 kHz to 4 kHz, sensitivity deviation ±10% at 23ºC 

ASDRA1K Damped Acceleration Transducer for Crash Test 
The ASDRA is an oil damped type acceleration transducer for crash tests with exchangeable cable. It is least affected from resonance. Rated Capacity: ±9807 m/s^{2} Dimensions: 5.4 x 5 x 13 mm Rated Output: 2 mV/V or more Frequency Response: DC to 1 kHz (Sensitivity deviation±3%), 1 k Hz to 4 kHz (Sensitivity deviation ±10% at 23ºC) 

ASM200BA Miniature Semiconductor Acceleration Transducer 
The ASM200BA is an undamped type semiconductor acceleration transducer for crash tests. Compact & lightweight ensure minimal influence to the vibration mode of a measuring object. Rated Capacity: 1961 m/s^{2} Dimensions: 7.6 x 4.6 x 15 mm Rated Output: 1.3 mV/V or more Frequency Response: DC to 3.5 kHz, sensitivity deviation: ±5% at 23ºC 

ASM1KBCV M3 Semiconductor Acceleration Transducer for Crash Test 
The ASM1KBCV M3 is an undamped type semiconductor acceleration transducer for crash tests. Compact & lightweight ensure minimal influence to the vibration mode of a measuring object. The sensitivity direction is perpendicular to the mounting surface. Rated Capacity: ±9807 m/s^{2} Dimensions: 12 x 5.4 x 25 mm Rated Output: 1.75 mV/V or more Frequency Response: DC to 2.8 kHz, sensitivity deviation: ±5% at 23ºC 

ASMAID Semiconductor Acceleration Transducer for Crash Test 
The ASMAID is a semiconductor acceleration transducer for crash tests. Rated Capacity: ±19613 m/s^{2} (2000 G) Rated Output: 6 mV/V or more Frequency Response: DC to 2.8 kHz, sensitivity deviation: ±5% at 23ºC 

ASM1KBCH M3 Semiconductor Acceleration Transducer for Crash Test 
The ASM1KBCH M3 is an undamped type semiconductor acceleration transducer for crash tests. Compact & lightweight ensure minimal influence to the vibration mode of a measuring object. The sensitivity direction is parallel to the mounting surface. Rated Capacity: ±9807 m/s^{2} Dimensions: 12 x 5.4 x 25 mm Rated Output: 1.75 mV/V or more Frequency Response: DC to 2.8 kHz, sensitivity deviation ±5% at 23ºC 

ASM1KCBV Semiconductor Acceleration Transducer for Crash Test 
The ASM1KCB is a undamped type semiconductor acceleration transducer for crash tests. Compact & lightweight ensure minimal influence to the vibration mode of a measuring object. The cable is exchangeable. Rated Capacity: ±9807 m/s^{2} Dimensions: 16 x 7.7 x 12.5 mm Rated Output: 1.5 mV/V or more Frequency Response: DC to 3.5 kHz, sensitivity deviation: ±5% at 23ºC 

LCF2TA Femur Load Cell 
For Femur of a Crash Dummy Mounted to the femur of a crash dummy such as Hybrid III, the LCF2TA detects impactinitiated load to the femur. 

LBTE Seat Belt Tension Transducer 
Downsizing, lightweight,and improved nonlinearity Even more compact and lightweight than former models. Additionally, the output correction circuit is builtin, with improved nonlinearity. This has 2 types  left and right depending on the cable direction. Types that support cable replacement using connectors can also be manufactured. 

LBTA20KNSA1 Seat Belt Tension Transducer 
Measuring ImpactInitiated Tesion of Seat Belt in Crash Tests Using a strain gage, the LBTS20KNS1 detects the impactinitiated tension applied through the dummy to the seat belt. 

LBTD100NSA1H Seat Belt Tension Transducer for CRS 
To use for measuring tensile force to install CRS (Child Restraint Seat) Measure the sensor of seat belt tension. Compact, lightweight and disassemblyfriendly design. 

LBTB200NSA1P Seat Belt Tension Transducer for CRS 
Compact, Lightweight, and Easy to Install and Remove Designed to measure the belt tension to fix the child seat, the LBTB200NSA1 is compact, lightweight and designed to be easily attached to, or removed from the belt. 

LSME15KNS1 Neck 6component Force Transducer 
Neck 6Component Force Transducer for Dummy Neck Upper Parts Mounted to the neck of a crash dummy such as Hybrid III, the LSME5KNS1 simultaneously detects 6 force components acting to the neck by impactinitiated inertia force generating in the head. 

AMAA Smallsized Triaxial Accelerometer 
The AMAA is a compact & lightweight accelerometer with large overloads resistant. Rated Capacity: ±19.61 to ±1961 m/s^{2} Dimensions: 10 x 5.8 x 14 mm (±19.61, ±49.03 m/s^{2}), 11 × 6.5 × 15 mm (±98.07 m/s^{2}), 19.3 × 11 × 11 mm (±980.7 m/s^{2}, ±1961 m/s^{2}) Rated Output: 2 V ±10% Frequency Response: DC to 500 Hz (±19.61, ±980.7, ±1961 m/s^{2}), DC to 200 Hz (±98.07 m/s^{2}) Safe Overloads: ±19613 m/s^{2} (±19.61, ±49.03 m/s^{2}), ±9807 m/s^{2} (±98.07, ±490.3 m/s^{2}), ±39227 m/s^{2} ±980.7, ±1961 m/s^{2}), 

ASPAA Piezoelectric Acceleration Transducer (Builtin Amplifier) 
The ASPAA is a piezoelectric acceleration transducer with builtin amplifier for measuring high range from slight to large acceleration in high sensitivity. Rated Capacity: ±2200 m/s^{2} Dimensions: ϕ7.8 x 15.7 x Hex7 mm Voltage Sensitivity: 1.0 mV/m/s^{2} ±10% Power Supply: 15 to 25 VDC, 0.5 to 5.0 mA 

ASPBA Piezoelectric Acceleration Transducer (Builtin Amplifier) 
The ASPBA is a piezoelectric acceleration transducer with a builtin amplifier for measuring high range from slight to large acceleration in high sensitivity. Rated Capacity: ±2200 m/s^{2} Dimensions: ϕ8.5 x 11.5 x Hex9 mm Voltage Sensitivity: 1.0 mV/m/s^{2} ±10% Power Supply: 15 to 25 VDC, 0.5 to 5.0 mA 

ASPCA Piezoelectric Acceleration Transducer (Builtin Amplifier) 
The ASPCA is a piezoelectric acceleration transducer with a builtin amplifier for measuring high range from slight to large acceleration in high sensitivity. Rated Capacity: ±360 to ±4000 m/s^{2} Dimensions: 14.2 x 14.2 x 23.7 mm Voltage Sensitivity: 1.0 mV/m/s^{2} ±10% or 10 mV/m/s^{2} ±10% Power Supply: 21 to 24 VDC, 0.5 to 10 mA 

DLTAS Inductancetype Displacement Transducer 
The DLTAS is an inductancetype displacement transducer with less friction and small measuring force. Rated Capacity: ±5 to ±500 mm Nonlinearity: Within ±0.5% RO Rated Output: Approx. 2 mV/V 

DLTBS Inductancetype Displacement Transducer 
The DLTBS is a waterproof displacement transducer. Rated Capacity: ±5 to ±500 mm Nonlinearity: Within ±0.5%RO Rated Output: Approx. 2 mV/V 

DTD Dial Gageequipped Displacement Transducer 
The DTD is a displacement transducer equiped with dial enables to visual measured displacement. Rated Capacity: 10, 20, 30, 50 mm Nonlinearity: Within ±0.5% RO Measuring Force: Approx. 1.7 N (10 mm), approx. 2.1 N (20 mm), approx. 2.2 N (30 mm), approx. 2.7 N (50mm) Rated Output: 1.5 mV/V or more 

DTHA Displacement Transducer 
The DTHA is a high accuracy, compact & lightweight displacement transducer. Rated Capacity: 5 to 100 mm Nonlinearity: Within ±0.1% RO Measuring Force: Approx. 1.5 to 4 N Rated Output: 5 mV/V ±0.1%, ±0.15% (5 mm) 

DTKA Displacement Transducer 
The DTKA is a smallsized displacement transducer for both tension and compression measurement. A measuring scale provided. Rated Capacity: 30, 50 mm Nonlinearity: Within ±0.3% RO Measuring Force: Approx. 2 N Rated Output: 2.5 mV/V or more The transducer can be measured in twoway tension and compression. 

GSATA900 Three Axis Angular Rate Gyro 
The GSATA is a compact & lightweight angular rate gyro for measurement of angular rate of 3 axes. Rated Capacity: ±15.71 rad/s (±900 deg/s ) Dimensions: 10 x 8 x 20 mm Rated Output: Approx. ±2 V Recommended Excitation: DC ±2.5 to 5.0 V with dual supply Safe Shock Resistant: 9807 m/s^{2} 

LCV Smallsized Compression Load Cell 
The LCV is a compact & lightweight load cell for measuring compressive loads. It is easy installation into existing facilities. The Steady Brace ensures to cancel transverse loads resulted from horizontal displacement. Rated Capacity: 50 kN to 200 kN Diameter: 68 mm (50 kN), 78 mm (100 kN), 98 mm (200 kN) Nonlinearity: Within ±0.05% RO 

LCHF Highaccuracy Compression Load Cell 
The LCHF is a high accuracy load cell. It is the highest measuring accuracy in high humidity place. Rated Capacity: 100 kN, 200 kN Diameter: 156 mm (100 kN), 176 mm (200 kN) Nonlinearity: Within ±0.02% RO 

LCKA Thin Compression Load Cell 
The LCKA is a compressive load cell with excellent accuracy, reliability, stability, and response. They also have a thin design for convenient installation as detection terminals of weighing systems. Rated Capacity: 5 kN to 200 kN Diameter: 78 mm (5 kN to 20 kN), 98 mm (50 kN), 108 mm (100 kN), 118 mm (200 kN) Nonlinearity: Within ±0.2% RO for 5 kN to 100 kN, within ±0.5% RO for 200 kN 

LCNA Smallsized Compression Load Cell 
The LCNA is a stainlesssteel load cell, for measuring compressive loads. It is compact & lightweight. Excellent stability and reliability are ensured by the hermeticallysealed structure with inert gas filled in. Rated Capacity: 500 N to 20 kN Diameter: 50 mm Nonlinearity: Within ±0.15% RO 

LCRGSA2 Smallsized Largecapacity Compression Load Cell 
The LCRGSA2 is a compact & lightweight load cell for measuring compressive loads. Rated Capacity: 10 to 50 kN Diameter: 20 mm (10 to 30 kN), 25 mm (50 kN) Nonlinearity: Within ±1% RO Others: Since the cable comes from the bottom, the load cells can be inserted into cylindrical objects for measurement. 

LCXAID Smallsized Compression Load Cell 
The LCXAID is a stainlesssteel load cell, for measuring compressive loads. It is compact & lightweight. Rated Capacity: 500 N to 20 kN Diameter: 28 mm (500 N to 2 kN), 43 mm (5 kN to 20 kN) Nonlinearity: Within ±0.1% RO 

LFMA Compact 6component Force Transducer 
The LFMA is a 6component force transducer. It enables simultaneous measurement of 3 components of force (Fx, Fy, Fz) in 3 axial directions and 3 moments (Mx, My, Mz) around the axes. Rated Capacity: 1 kN, 3 kN Diameter: 66 mm Nonlinearity: Within ±0.5% RO 

LFXA Compact 6component Force Transducer with Builtin Amplifier 
The LFXA is a 6component force transducer with builtin Amplifiers. It enables simultaneous measurement of 3 components of force (Fx, Fy, Fz) in 3 axial directions and 3 moments (Mx, My, Mz) around the axes. Hollowed structure for wiring easily. Rated Capacity: 1 kN, 3 kN Diameter: 66 mm Nonlinearity: Within ±0.5% RO 

LMAA Smallsized Compression Load Cell 
The LMAA is a compact & lightweight load cell for measuring compressive loads. Rated Capacity: 5 N to 1 kN Diameter: 12 mm (5 to 50 N), 20 mm (100 N to 1 kN) Nonlinearity: Within ±1% RO 

LMBA Smallsized Compression Load Cell 
The LMBA is an utrasmall load cell for measuring compressive loads. Rated Capacity: 10 N to 2 kN Diameter: 10 mm (10 to 200 N), 16 mm (500 N to 2 kN) Nonlinearity: Within ±0.5% RO 

LMBTA Smallsized Compression Load Cell 
The LMBTA is an ultrasmall load cell for measuring compressive loads. Its operating temperature is up to 100ºC. Rated Capacity: 50 N to 2 kN Diameter: 10 mm (50 to 200 N), 16 mm (500 N to 2 kN) Nonlinearity: Within ±0.3% RO 

LMCA Smallsized Compression Load Cell 
The LMCA is a compact & lightweight load cell for measuring compressive loads. Rated Capacity: 5 kN to 50 kN Diameter: 12±0.1mm （5, 10 kN), 16±0.1mm (20 kN), 22±0.2mm (50 kN) Nonlinearity: Within ±0.5% RO for 5 kN to 10 kN, within ±1% RO for 20 kN to 50 kN 

LMRSSA2 Smallsized Compression Load Cell 
The LMRSSA2 is a compact & lightweight load cell for measuring compressive loads. Rated Capacity: 2 kN to 20 kN Diameter: 21 mm Nonlinearity: Within ±1% RO, within ±2% RO for 20 kN 

LSMBSA1 3component Force Transducer 
The LSMBSA1 is a strain gage based 3component force transducer for simultaneously measuring force in 3 directions of X, Y, and Z. Rated Capacity: 10 to 500 N Diameter: 88 mm Nonlinearity: Within ±0.5% RO 

LUE Tension/Compression Load Cell 
The LUE is a load cell for measuring tension or compression loads. It is hermeticallyseal structure with inert gas filled in. Rated Capacity: ±500 N to ±200 kN Diameter: 80 mm (±500 N to ±2 kN), 100 mm (±5 kN to ±20 kN), 112 mm (±50 kN), 138 mm (±100 kN), 186 mm (±200 kN) Nonlinearity: Within ±0.2% RO 

LUHF Highaccuracy Tension/Compression Load Cell 
The LUHF is a load cell for measuring tension or compression loads. It is hermeticallyseal structure with inert gas filled in and excellent in zero float characteristics. Rated Capacity: ±500 N to ±200 kN Diameter: 100 mm (±500 N to ±5 kN), 100 mm (±10 kN to ±20 kN), 130 mm (±50 kN), 160 mm (±100 kN), 200 mm (±200 kN) Nonlinearity: Within ±0.02% RO 

LUKA Tension/Compression Load Cell 
The LUKA is a a thin load cell for measuring tension or compression loads. It can be installed to where the height is limited. It also suitable for fatigue test where repeativite loads are applied continuously. Rated Capacity: ±5 kN to ±2 MN Diameter: 77 mm (±5 kN to ±10 kN), 107 mm (±20 kN), 127 mm (±50 kN), 157 mm (±100 kN), 227 mm (±200 kN), 307 mm (±500 kN), 375 mm (±1 MN), 560 mm (±2 MN) Nonlinearity: Within ±0.1% RO for 5 kN to 200 kN, within ±0.2% RO for 500 kN to 2 MN 

LURASA1 Compact Tension/Compression Load Cell 
The LURASA1 is a load cell for measuring tension or compression loads. Rated Capacity: ±50 N to ±2 kN Diameter: 28 mm Nonlinearity: Within ±0.5% RO 

LURBSA1 Jack Load Cell 
The LURBSA1 is load cell for measuring loads applied to jacks. Rated Capacity: ±10 kN to ±2 MN Diameter: 100 mm (±10 kN), 110 mm (±20 kN), 125 mm (±30 kN, ±50 kN), 175 mm (±100 kN), 255 mm (±200 kN, ±300 kN), 330 mm (±500 kN), 430 mm (±1 MN), 530 mm (±1.5 MN), 590 mm (±2 MN) Nonlinearity: Within ±0.2% RO for 10 kN to 200 kN, within ±0.5% RO for 300 kN to 2 MN 

LUXBID Compact Tension/Compression Load Cell 
The LUXBID is a compact stainlesssteel load cell with waterproof connector for measuring tension or compression loads. Models for withstand to repeated bending (Flexible cable) are available. Rated Capacity: ±50 N to ±20 kN Diameter: 43 mm Nonlinearity: Within ±0.1% RO for 5 kN to 20 kN, within ±0.15% RO for 50 N to 2 kN 

PGH High Pressure Transducer 
The PGH is a high pressure transducer with hermetically sealed with inert gas filled in. It is suitable for high pressure measurement. Rated Capacity: 100, 200 MPa Compensated Temperature: 10 to 70ºC 

PGU Pressure Transducer 
The PGU is a pressure transducer with high accuracy and reliability because of hermeticallysealed structure with inert gas filled in. Rated Capacity: 200 kPa to 50 MPa Compensated Temperature: 10 to 60ºC 

PGLA Smallsized Pressure Transducer 
The PGLA is a compact & lightweight pressure transducer with the semiflush diaphragm at the end. It is high accuracy. Rated Capacity: 1 to 50 MPa Compensated Temperature: 10 to 60ºC 

PGMD Smallsized Pressure Transducer 
The PGMD is a smallsized pressure transducer with the diaphragm at the end. Rated Capacity: 5 to 50 MPa Compensated Temperature: 0 to 60ºC 

PGME Smallsized Pressure Transducer 
The PGME is a smallsized pressure transducer with the semiflush diaphragm at the end. Rated Capacity: 1 to 50 MPa Compensated Temperature: 10 to 60ºC 

PGMH Smallsized Pressure Transducer 
The PGMH is a smallsized pressure transducer with the semiflush diaphragm at the end. Rated Capacity: 500 kPa to 50 MPa Compensated Temperature: 10 to 60ºC 

PGMCA Smallsized Pressure Transducer 
The PGMCA is a compact pressure transducer with flushdiaphragm at the end. Rated Capacity: 200 kPa to 1 MPa Compensated Temperature: 0 to 50ºC 

PGSA Smallsized Pressure Transducer 
The PGSA is a compact & lightweight pressure transducer. Rated Capacity: 1 to 50 MPa Compensated Temperature: 10 to 60ºC 

PHBA High/Lowtemperature Pressure Transducer 
The PHBA is a pressure transducer for measuring pressure at high or low temperatures. The sensing surface is a stainless steel diaphragm. Rated Capacity: 1 to 50 MPa Compensated Temperature: 196 to 200ºC 

PHCB Flush Diaphragm Type Hightemperature Pressure Transducer 
The PHCB is a flushdiaphragm type pressure transducer for measuring pressure at high temperature. It adopt thinfilm strain gage formed by sputtering. Rated Capacity: 2 to 20 MPa Compensated Temperature: 23 to 230ºC 

PHFSS3 Series Smallsized Hightemperature Pressure Transducer 
Excellent in environmental performance Rated Capacity: 2 to 20 MPa Compensated Temperature: 40 to 150°C 

PHLA Smallsized High/Lowtemperature Pressure Transducer 
The PHLA is a compact & lightweight pressure transducer with the semiflush diaphragm at the end. It is suitable for measurement under high or low temperature. Rated Capacity: 1 to 50 MPa Compensated Temperature: 20 to 150ºC for PHLAA, 196 to 200ºC for PHLAB 

PSC Miniature Pressure Sensor 
The PSC is a miniature pressure sensor with a straingage bridge inside. The cable direction is parallel to the sensor surface. Rated Capacity: 50 kPa to 7 MPa Compensated Temperature: 0 to 50ºC 

PSD Miniature Pressure Sensor 
The PSD is a miniature pressure sensor with a straingage bridge inside. The cable direction is at right angle to the sensor surface. Rated Capacity: 50 kPa to 7 MPa Compensated Temperature: 0 to 50ºC 

PSS Miniature Pressure Sensor 
The PSS is a miniature pressure sensor with a straingage bridge inside. Thinfilm strain gages are directly formed on the diaphragm. Rated Capacity: 20, 50, 100 kPa Compensated Temperature: 0 to 50ºC (As for PSSAF/BF, Noncondensing) 

TPHA Highly Rigid Torque Transducer 
The TPHA is a noncontact highly rigid torque transducer with high frequency response and high accuracy. Rated Capacity: ±500 N･m to ±50 kN･m Rated Output: ±10 V, 4 to 20 mA Max. Rotation Speed: 10000 rpm for ±500 N･m, 5000 rpm for ±10 kN･m, and 3000 rpm for ±20 to 50 kN･m 

TPSA Builtin Amplifier Torque Transducer 
The TPSA is a compact & lightweight torque transducer with buitin amplifier. Easy maitenance due to noncontact transmission. Rated Capacity: ±0.1 N･m to ±5 kN･m Rated Output: ± (5+0.2) V at load resistance 5 kΩ or more, 2 kΩ or more for ±0.11 N･m to ±2 N･m Max. Rotation Speed: 15000 rpm for ± 0.1 N・m to ±2 N・m, 5000 rpm for ±5 N･m to ±500 N･m, 4000 rpm for ±1 kN･m, and 3000 rpm for ±5 kN･m 

LBRS1KNSA2 Hand Brake Operating Force Transducer 
The LBRS is a load cell to measure tension of parking brakes. It can be install to a parking brakes easily. Thetransducer can measure vertical force applied to the parking brake accurately. 