Upgrade history NTB-500A
- TEDS information for the measuring unit 1 was read in the measuring unit 2. This failure occurred in reading TEDS information with TEDS-installed sensors connected to the measuring unit 1 and no TEDS-installed sensor, to the measuring unit 2. If the measuring unit 2 has any TED-installed sensor connected, TEDS is normally read in the channel. Also, this failure can be eliminated by unchecking the setting on TEDS information window of DCS-100A.
- Writing user data cleared TEDS user data. This failure occurs with the channels for which DCS-100A user data is not revised. Revising the user data enables correct data writing
The following failures are corrected: